gmi prosthetic kit consists of an autoclavable case made of high strength resin, containing all the necessary material for the correct insertion of the whole range of attachments of the gmi implant systems. In addition, the kit includes the abutment extractors for the Avantgard system.
Case made of high thermal and impact strength technical resin.
Security lid locking avoiding accidental opening.
Inner supports made of high quality colored silicon.
Extractable inner tray with legs to make easier the manipulation.
Clear and natural components distribution.
Color coded silicones according to implant system for they was designed:
Blue: Avantgard abutments extraction wrenches.
Grey: common instruments.
Free socket that allow storing optional instrumental.
1. Cleaning – Removal of organic material should be performed in an ultrasound basin starting at 24.000 Hz, using enzyme soap (at least 3 enzymes) at the dilution recommended by the manufacturer. Wash with running water and perform a visual check to detect the presence of waste. If positive, repeat the process and should it persist, eliminate by mechanical cleaning. The sterilizations boxes must be cleaned before each use.
2. Conditioning for sterilization – Condition the product on a surgical grade compatible size paper envelope.
3. Moist heat sterilization – shall be conducted in an autoclave without exceeding the maximum values set: 134ºC, 20 minutes, 2.2 bar. Warning – For the sterilization procedure, carefully follow the recommendations in the instruction manual of the autoclave manufacturer.
4. Product life – The useful life of the box will depend on the treatment received during its use.
5. Important notes:
GMI surgical kits are not supplied in sterile condition and then the user must provide for sterilization before use.
Exclusive model for autoclave sterilizing.
Clean and sterilize before use.
Do not sterilize in an oven as this can cause the box to lose its shape or melt.
Avoid the box coming into contact with chemical products as they could cause cracking, staining or general degradation of the product.
Carefully follow the guidelines given in the instruction manual of the autoclave manufacturer.